Analytical Chemistry & Materials Development Group at NASA JPL, Pasadena, CA
ionRocket visualizes hidden compounds in your samples. IonRocket generates a temperature gradient from ambient up to as high as 600ºC over several minutes. This allows compounds in your samples to be sublimated, vaporized, or pyrolized according to their volatility, and then introduced into the DART® gas stream.
Data obtained from ionRocket yields another axis of data (time/temperature) beyond that obtained from normal DART®-MS (Direct Analysis in Real Time-Mass Spectrometry) analysis. Species desorb in order of their volatility along the temperature gradient, and thus are separated in time. The data thus resembles that of an LC-MS or GC-MS chromatogram, consisting of temperature/time, m/z, and intensity. This can separate rare from abundant species, making them easier to detect.
US Patent 8,927,926
Yamaguchi, M. (2020). Thermal desorption and pyrolysis direct analysis in real time mass spectrometry of Nafion membrane. Journal of Applied Polymer Science
Bridge, C. & Marić, M. (2019). Temperature-Dependent DART-MS Analysis of Sexual Lubricants to Increase Accurate Associations. J. Am. Soc. Mass Spectrom.
Barnett, I., Bailey, F. C., & Zhang, M. (2019). Detection and Classification of Ignitable Liquid Residues in the Presence of Matrix Interferences by Using Direct Analysis in Real Time Mass Spectrometry. Journal of Forensic Sciences.
Cody, R. B., & Fouquet, T. (2018). “Reverse Kendrick Mass Defect Analysis”: Rotating Mass Defect Graphs to Determine Oligomer Compositions for Homopolymers. Analytical chemistry, 90(21), 12854-12860.
Price, E. R., McClure, P. J., Jacobs, R. L., & Espinoza, E. O. Identification of rhinoceros keratin using direct analysis in real time (DART) time‐of‐flight mass spectrometry (TOFMS) and multivariate statistical analysis. Rapid Communications in Mass Spectrometry.
Abe, H., Takei, C., Sakakura, M., Yajima, D., & Iwase, H. (2018). Comprehensive Drug Screening by Thermal Desorption and Pyrolysis Combined with Direct Analysis in Real Time-Mass Spectrometry (TDP/DART-MS). Analysis of Drugs of Abuse (pp. 115-124). Humana Press, New York, NY.
M Maric, J Marano, RB Cody, C Bridge (2018). DART-MS: A New Analytical Technique for Forensic Paint Analysis. Analytical chemistry
|Main Body Outer Dimension / Weight||315 mm (W) x 120 mm (D) x 175 mm (H) / 3.2 kg|
|Heater Control Box Dimension / Weight||321 mm (W) x 220 mm (D) x 147 mm (H) / 4.1 kg|
|Heating Control Temperature||Room temperature – 600°C|
|Main Power||100 VAC/240 V, 50 Hz/60 Hz|
|Operating Temperature Environment||10°C to 40°C (No condensation)|
|Power Consumption||During operation: 100 VAC 320 W
As stand-by: 100 VAC 32 W
We can send you more information or run one of your samples.